---
product_id: 22004588
title: "Flowline DX10-01 EchoPod Ultrasonic Level Sensor with Fob, 1\" NPT"
brand: "flowline"
price: "$30357"
currency: MXN
in_stock: true
reviews_count: 1
url: https://www.desertcart.mx/products/22004588-flowline-dx10-01-echopod-ultrasonic-level-sensor-with-fob-1
store_origin: MX
region: Mexico
---

# Corrosion-resistant 49.2" range WebCal config Flowline DX10-01 EchoPod Ultrasonic Level Sensor with Fob, 1" NPT

**Brand:** flowline
**Price:** $30357
**Availability:** ✅ In Stock

## Summary

> 🚀 Elevate Your Measurement Game!

## Quick Answers

- **What is this?** Flowline DX10-01 EchoPod Ultrasonic Level Sensor with Fob, 1" NPT by flowline
- **How much does it cost?** $30357 with free shipping
- **Is it available?** Yes, in stock and ready to ship
- **Where can I buy it?** [www.desertcart.mx](https://www.desertcart.mx/products/22004588-flowline-dx10-01-echopod-ultrasonic-level-sensor-with-fob-1)

## Best For

- flowline enthusiasts

## Why This Product

- Trusted flowline brand quality
- Free international shipping included
- Worldwide delivery with tracking
- 15-day hassle-free returns

## Key Features

- • **Durable Design:** Constructed with a PVDF transducer and 6P polycarbonate enclosure, built to withstand corrosive liquids.
- • **User-Friendly Setup:** Experience hassle-free configuration with WebCal software and USB adapter—get started in no time!
- • **Precision Measurement:** Achieve continuous non-contact level measurement up to 49.2 inches, ensuring accuracy in every application.
- • **Optimized for Small Tanks:** Designed with a narrow 2" beam width and short 2" dead band, perfect for small tank applications.
- • **Smart Temperature Control:** Automatic temperature compensation guarantees reliable measurements, no matter the conditions.

## Overview

The Flowline DX10-01 EchoPod Ultrasonic Level Sensor is engineered for precision and durability, offering non-contact level measurement up to 49.2 inches. With user-friendly configuration via WebCal software, it's optimized for small tanks and built to resist corrosive environments, ensuring accurate readings with automatic temperature compensation.

## Description

The general purpose ultrasonic level transmitter provides continuous level measurement up to 49.2" (1.25m) with a selectable 0-5 VDC, 0-10 VDC or 976-2000 Hz frequency signal output, and is configured via our free WebCal software. Select the voltage output for interface with analog input cards. Select the frequency output for interface with discrete input cards. This non-contact liquid level sensor is ideally suited for corrosive, sticky or dirty liquids, and is widely selected for small day tank, skid, IBC, sump and process tank level applications. Range: 49.2" (1.25m). Accuracy: 0.125" (3mm). Resolution: 0.019" (0.5mm). Dead band: 2" (5cm). Beam width: 2" (5cm). Configuration: WebCal PC Windows USB 2.0. Memory: Non-volatile. Supply voltage: 12-28 VDC. Consumption: 0.5W. Signal output: 0-5 V, 0-10 V, 976-2000 Hz. Frequency: Sink current, 15 mA nominal. Signal invert: 5-0 V, 10-0 V, 2000-976 Hz. Signal fail-safe: Full, empty or hold last. Process temperature: F: 20 to 140d egree C: -7 to 60 degree. Temperature comp.: Automatic. Ambient temperature: -31 to 140 degree F, -35 to 60 degree C. Pressure: MWP = 30 PSI (2 bar). Enclosure rating: Type 6P, encapsulated,corrosion resistant and submersible. Enclosure material: Polycarbonate. Strain relief mat: Santoprene. Trans. material: PVDF. Cable jacket material: Polyurethane. Cable type: 6-conductor, shielded. Cable length: 48" (1.2m). Process mount: 1" NPT. Mount gasket: Viton. Classification: General purpose. Compliance: CE, RoHS. Approvals: cFMus.

## Features

- Continuous non-contact level measurement up to 49.2" (1.25m)
- Configuration is fast and easy via WebCal software and USB adapter
- Narrow 2" beam width and short 2" dead band optimized for small tanks
- PVDF transducer and 6P polycarbonate enclosure for corrosive liquids
- Automatic temperature compensation for accurate measurement

## Images

![Flowline DX10-01 EchoPod Ultrasonic Level Sensor with Fob, 1" NPT - Image 1](https://m.media-amazon.com/images/I/61HJMFqqz+L.jpg)

## Customer Reviews

### ⭐⭐⭐⭐⭐ 







  
    Producto de la mejor calidad
  
  

*by G***O on Reviewed in the United States 🇺🇸 on June 24, 2021*

Flowline me parece que es de muy buena calidad. En los proyectos que los he usado se ve su exactitud y buen funcionamiento. El software de configuración es muy bueno.

---

## Why Shop on Desertcart?

- 🛒 **Trusted by 1.3+ Million Shoppers** — Serving international shoppers since 2016
- 🌍 **Shop Globally** — Access 737+ million products across 21 categories
- 💰 **No Hidden Fees** — All customs, duties, and taxes included in the price
- 🔄 **15-Day Free Returns** — Hassle-free returns (30 days for PRO members)
- 🔒 **Secure Payments** — Trusted payment options with buyer protection
- ⭐ **TrustPilot Rated 4.5/5** — Based on 8,000+ happy customer reviews

**Shop now:** [https://www.desertcart.mx/products/22004588-flowline-dx10-01-echopod-ultrasonic-level-sensor-with-fob-1](https://www.desertcart.mx/products/22004588-flowline-dx10-01-echopod-ultrasonic-level-sensor-with-fob-1)

---

*Product available on Desertcart Mexico*
*Store origin: MX*
*Last updated: 2026-03-09*